High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||15 October 2015|
|PDF File Size:||8.29 Mb|
|ePub File Size:||18.38 Mb|
|Price:||Free* [*Free Regsitration Required]|
Got it, continue to print. The green control lamps 3 show which mode is activated. General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, hermke be operated in explosive-hazardous locations.
G131038 Hermle Z 233 M-2 Labnet Centrifuge W/rotor
There is a distinction between two different kinds of errors. Overcurrent or undervoltage due to power supply fluctuations Action Page 14 – Keyboard – Starting the centrifuge – “qu Hermel available tubes or bottles made of glass or plastic. Preselection Of Operating Time 4.
By pressing ‘print’ button you will print only current page. Never hermlee in the safety zone of 30 cm around the centrifuge or deposit dangerous materials inside this zone during centrifugation.
Table of Contents Add to my manuals Add. Operation Operation Operation Operation Operation 4. If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm.
Don’t have an account? The digital indications on the display will light up when the power is on. Turn the speed knob 1see figure 4 till the required rotor code hermld on the d i s p l a y. Power failure while rotor is in motion Action: Page 20 Error no.: Use only original spare parts! The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations.
The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, z2333 time and the speed of the rotor.
HERMLE Z M-2 INSTRUCTION MANUAL Pdf Download.
Service And Inspection Of Centrifuge 7. Enter text from picture: Width 28 cm Depth 37 cm Height To open the centrifuge lid. Comments to this Manuals Your Name.
If this hemle not done, they could scratch the protective coating of the rotor. Description – The rotor is not installed securely on the motor shaft. Page of 20 Go.
Figure 8 If the green “fast” control lamp is lit, the unit accelerates and decelerates fast. Page 19 Error no.: Don’t show me this message again. The rotor must always be balanced in this fashion prior to operation. Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Should this happen, the centrifuge, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately.
To print the manual completely, please, download m–2. Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other. During a run and with the centrifuge lid closed you have to press the “preset” key 1hold it and change operating time by turning the knob 3. On the digital display “time min-sec” the letters “ro” rotor appears.
Page 15 – Temperture Page 16 – Service and inspection of centrifuge Page 17 – Cleaning of centrifuge after breakage of Open and reclose the lid, heemle the centrifuge Error no.: With the centrifuge lid open you can preset the operating time with the knob 3. Keyboard – Starting The Centrifuge – “quick”-key 4.